The offered microscope must be of latest technology and of special ergonomic design, suitable for preparation observation in br9ght-field, with extension possibility to further microscopy techniques like dark field, pollarisation etc.
The image information should take digital form directly from the camera head. The camera software should integrate the functions of the microscope, the digital camera and the software in one work interface facilitating the digital imaging procedure. The digital shooting of microphotos must be automated through the microscope functions.
Must allow the projection, adjustment and storage of microscopy images in an archive library (thumbnail gallery), archiving images with comments, measurements and operational data of the microscope It should have functions of scale installation, length measuring, commenting, pointers, headers and color editing in the stored images as well as projection of live image and adjustment of shooting parameters. Must provide comparing between live and stored images or two stored ones.